# Pulsed neutron generators MFNG-601

Main technical specifications of **pulsed neutron generator MFNG-601 (МФНГ-601**), applied in borehole tool PILK-76(ПИЛК-76) to solve reservoir permeability and porosity determination tasks and also in tools APILK-160 (АПИЛК-160), APILK-102 (АПИЛК-102), APILK-90 (АПИЛК-90) and PIMS-76 (ПИМС-76).

Parameter | Value | Note |

Diameter, length, mm | 60, 1200 | |

Supply voltage, V | DC positive, 18÷30 | Selected depending on the applied borehole instrumentation – self-contained or borehole |

Electric energy efficiency rate, W/neutron | 16×10^{8} |
Defined as electric energy, consumed by the whole neutron generator per one neutron (at output of 10^{8} n/sec consumed power is 16 W) |

Neutron maximum frequency of oscillation, Hz | 20000 | |

Neutron minimum frequency of oscillation, Hz | Continuous radiation | Practically frequencies lower than 400 Hz are not applied. |

Maximum ambient temperature, °C | 150 | |

Timing capability | Internal / external | |

Neutron pulse leading / drop edge, μs | 2/1 | |

Neutron pulse position stability, μs | ±0.4/0.7 |